|Minimum Order Quantity||1 Unit|
|Brand||PRAGATI PROCESS CONTROLS|
|Output Voltage||a) Auxiliary AC Voltage:Op ranges: 65 - 130 - 260 V b)Auxiliary DC Op Ranges: 130 V and 240 V.|
|Usage/Application||suited for the testing and adjustments of all the types of production and distribution relays|
|Country of Origin||Made in India|
Secondary Injection Relay Testing Kit
The single phase relay test set should be suited for the testing and adjustments of all the types of production and distribution relays. In addition to the above, it should also test converters and energy meters.
Mains supply: 220 V ± 15%; 50-60 Hz.
The main generator shall have three outputs: currents, voltage AC, voltage DC with provision of generating at maximum power or at reduced power.
On all outputs the capability of - Current Ranges: 10, 40, 100 A. Output power: 800 VA. AC voltage range: 250 V. Output power: 500 VA.
DC voltage range: 300 V. Output power: 300 W.
Auxiliary AC Voltage
Output ranges: 65 – 130 - 260 V. Output power: 30 VA.
Auxiliary DC Voltage
Ranges: 130 V and 240 V.
Power: 90 W at full range; max. currents: 0.9 A @ 130 V and 0.45 A @ 240 V.
The timer shall have two inputs, START and STOP.
Inputs level: Normal Open or Normal Close. Clean or under voltage, maximum 275 V DC. Timer ranges: 9.999 s; 99.99 s; 99999.9 s. Accuracy: ± 0.005% of value ± 0.01% of range.
Two auxiliary make and break contacts shall have a delay from 0 to 99.99 s with respect to test start. Contacts characteristics: 5 A; 250 V AC; 120 V DC
The display: AC current (AC or DC voltage); auxiliary AC voltage; auxiliary DC voltage. Accuracy, DC: 0.5% of reading ± 0, 1% of range; AC: 1% of reading ± 0,2% of range.
External Input Measurement
Input current ranges. 20 mA or 10 A, AC or DC. Accuracy, DC: 0.5% of reading ± 0,1% of range; AC: 1% of reading ± 0,2% of range. Maximum input voltage: 600 V, AC or DC.
Selections, for the test of over- or under-(current, voltage.) relays. . OFF: main outputs not generated;. ON: all outputs generated;. From OFF to ON + time: all outputs shall be generated and timer shall start; at STOP, outputs are removed and time shall be displayed. . From ON to OFF + time: outputs shall be removed and timer shall start; at STOP, the time shall be displayed. - Reclose test.